A guide to selecting the right float switch for your application
By Simon Dear of Cynergy3 Components
When selecting a float switch for a liquid level sensing application, a number of factors need to be considered, including float switch and gasket materials, physical arrangement, electrical ratings and cable type, says Simon Dear of Cynergy3 Components.
Many industrial processes require devices that are able to sense the level of liquid stored within various types of tank or chamber. The signals from these devices may be used to control the production process or to provide indication of the status.
One of the most reliable, well-proven technologies for liquid level sensing is a float switch. This type of switch comprises a magnet contained within a float, as well as a magnetic reed switch contained within a fixed housing. The movement of the float, due to the changing liquid level, will cause the reed switch to operate (i.e. close or open) at a particular level. This tried and trusted technology is based on a relatively simple design that offers long term reliability without the need for the user to calibrate the switch.
Whilst there are different methods available for selecting the right float switch for a particular application, the main factors to consider include the following:
Physical arrangement and style
The choice of styles that may be suitable for an application will depend on the physical arrangement of the tank, the available mounting positions and whether access is available to the inside of the tank. The main styles are horizontal/side mounting and vertical mounting. The horizontal/side mounting type normally has a fixed housing, which passes through the sidewall of a tank, with a hinged float attached to the fixed housing. Vertical mounting types normally have a fixed vertical stem, which is installed through the top or bottom of a tank, along which slides a cylindrical float.
Another important consideration is whether a build up of deposits from the liquid on the float body is likely to occur. These deposits can, over a period of time, accumulate to such an extent that the float switch can fail to operate. Whilst particular types of float switch have been developed to limit the effects of this build up, the experience and knowledge of the switch manufacturer is also critical here.
It is critical to select a float switch that is constructed from the right materials that are compatible with the liquids and temperatures of the particular application. Component damage as a result of incorrect materials selection can ultimately cause failure of a float switch, which may have severe consequences. Typical float switch materials include:
Nylon: suitable for many oils, diesel, organic chemicals and MEK-based printing inks.
Polypropylene (PP): suitable for many acids and alkali, detergents, inorganic and organic chemicals, oils and water.
Polyphenylene sulphide (PPS): suitable for many of the more aggressive chemicals and higher process temperatures, up to 120ºC.
Buna/Nitrophyl: suitable for many oils, diesel, petrol and water (non-potable applications).
D300 foam (PVC): suitable for most hydraulic oils and many chemical solutions.
Stainless steel: suitable for most medical and food applications, chemicals, hydraulic fluids, fuel oils and applications with process temperatures up to 135ºC.
Selection of the most suitable materials for both float switch and gasket can be made by referring to a ‘Chemical Compatibility’ table. These tables provide a good indication of the suitability of the various float switch materials in a wide range of liquids. For some process liquids, it may be necessary to obtain a sample float switch in order to test the compatibility.
It is important to fully understand the nature of the load that needs to be switched and to ensure that the float switch is capable of handling this load. The electrical ratings, which most manufacturers provide in their float switch specifications, are for purely resistive loads. Any loads that have either inductive or capacitive components should have the appropriate contact protection measure applied.
In applications where aggressive liquids may spill onto external wiring, it is important to specify particular materials for the cables used to connect to the float switches. There are standard, UL-approved cable types for the various float switches, as well as high temperature, low smoke zero halogen (LSZH) and other specialised materials.
Custom engineered float switches
The operating environment is critical to the choice of float switch. A water tank for an industrial process may only require a simple plastic float switch. However, if the application is in a hazardous area, for example, a petrochemical storage tank where flammable gases, vapours or dust are present, a stainless steel, explosion-proof float switch will be required.
Over the years, Cynergy3 has custom engineered many variants of its float switches to match particular customer requirements in a wide range if industry sectors including oil, gas & petrochemicals; food and beverage; chemicals and pharmaceuticals; water and wastewater; and process manufacturing. These include float switches for all types of environment, including industrial process control, safe areas, intrinsically safe and hazardous areas (ATEX-certified), as well as WRAS-approved switches for drinking water applications.
Cynergy3 is a manufacturer of liquid level and flow sensors, HV and RF relays. The company has developed an extensive range of liquid level sensors over the past 44 years by focusing on solutions for customer applications. Cynergy3 has been at the forefront of RF relay technology for 30 years, during which time it has developed relays for many specific communications applications, by working with world leading manufacturers of radio communications equipment.
The Dawn of the Programmable Logic Controller (PLC)
By Ken Ball
Programmable Logic Controllers (PLCs) were more of a technology evolution than a radical development. A noteworthy aspect about the...
Robotic Welding Advantages
By Tom Bonine, National Metal Fabricators
Robotic welding is highly cost-effective, especially for longer production runs where programming time...
The Architecture of the SCADA Mobility Infrastructure
By Edward Nugent, PcVue Inc., et al
The different way in which mobile device users interact with their devices compared to operator’s...
Expecting one-year ROI on automation systems is short sighted
By Bill Lydon, Editor
Many users, systems integrators, and manufacturing managers expect one year to eighteen month Return on Investment (ROI)...
Workflow and Automation for Account and Password Management
By Dean Wiech, Tools4Ever
Workflow and automation procedures for identity and password management can reduce time spent, increase security and...
Boulting upgrades pumps at 90 year-old pump station
The original pumps are connected to new high performance motors and gearboxes, close to the oldest static steam driven pump, which will be kept as...
Web seminar: How to calibrate temperature instruments
ISA and Beamex will co-host a free web seminar 1 September 2015 on how to calibrate temperature instruments.
NEMA says Business Conditions Rebounded Sharply in July
NEMA’s Electroindustry Business Conditions Index (EBCI) for current conditions in North America improved by more than 20 points, marking an...
Panduit Acquires IntraVUE Network Diagnostic Software
IntraVUE provides the capability to identify and report information critical to improving uptime of the Industrial Ethernet infrastructure.
Exida publishes Safety Equipment Reliablity Handbook - 4th Edition
The SERH database provides a collection of failure rate data that is applicable for use in Safety Instrumented System (SIS) conceptual design...