Direct versus Indirect measurement of shaft angle
By Mark Howard, Zettlex
As a rule, it’s preferable to measure directly the position or speed of the object that you are interested in. In many cases, practical problems of physical environment or limited space mean that this can be a challenge, particularly when measuring the angle of shafts with a diameter greater than a couple of inches. The traditional approach is to measure angular position or speed indirectly – typically inferring the shaft’s position from measurements made elsewhere. It can also be done with inductive sensors.
Traditional Approach – Indirect Measurement
Potentiometers, resolvers and optical encoders are the most common devices for measuring shaft angle. Potentiometers offer a simple, low cost solution but are unsuitable for harsh environments or continuous rotation. Resolvers are reliable in tough conditions but their high cost and bulk mean that they are rare outside the defence, aerospace, and oil & gas sectors. Optical encoders are not as robust as resolvers but are widely available and keenly priced. Most optical encoders have a small (typically <1/2” diameter) input shaft with incremental pulse output. Through-shaft versions are available but the bore is usually limited to <2”. Above this, encoder prices increase dramatically and availability dwindles.
So how do you measure the angle of a large diameter through-shaft of, say, 3” or larger? Traditionally, a smaller, secondary shaft is driven from the larger primary shaft and the angle of the secondary shaft is measured. In other words, the angle of the primary is measured indirectly or inferred. For many years, this has been the approach in gun turrets, rotary tables, radar antennae, security cameras, large motors, medical scanners and telescopes. As the secondary shaft is smaller, there is a wide choice of rotary encoders. If absolute (rather than incremental) angle of the primary shaft is required, then additional gearing or a multi-turn encoder can be used.
Problems with the Indirect Approach
The angle of the primary shaft is calculated from the angle of the secondary – assuming that their relative rotation is proportional. Not unreasonable? As ever, the devil is in the detail and, in practice, there are problems with this assumption.
As a general rule, if the required measurement accuracy is <1 degree, indirect measurement is probably not going to work reliably – or at least not for long. There are two parts to the problem – accuracy and reliability. Inaccuracy comes from the number of factors in the system’s tolerance stack up. For a system coupled by gears, these factors include, but are not limited to:
1. Encoder accuracy.
2. Encoder thermal coefficients – i.e. drift in output due to temperature.
3. Differential thermal expansion in gears, shafts, bearings, mounts, etc.
4. Gear backlash.
5. Gear wear.
6. Concentricity of gears on shafts.
7. Gear train/tooth strain versus torque.
8. Shaft concentricity.
9. Variation of gear position with shock or vibration.
10. Tolerance on gear tooth position.
11. Tolerance on primary and secondary shaft centres.
12. Variation in shaft centre distance - due to load/bearing clearances.
13. Variations from lubrication – due to amount, type and viscosity.
14. Mechanical friction – especially stiction.
15. Effect of foreign matter on gear teeth.
16. Twist due to torque in shafts.
17. Shaft bending.
Each of these effects alone is probably not a major influence on accuracy. The problem arises because all these effects stack together.
A common misconception in #1 is that an encoder with 1,000 counts per revolution is accurate to 1/1000th of a rev. Unfortunately, resolution is not the same as accuracy.
For reliability, most engineers know that the reliability of any mechanical system is proportional to the number of parts in it – especially moving parts. Gear, pulley or chain systems are susceptible to foreign matter. This can often be overlooked by engineers who expect their equipment will be operated within the specified envelope and that all servicing will be carried out by skilled personnel who always replace baffles and seals.
Experience shows this is wishful thinking. Foreign matter often arises from unexpected, sometimes bizarre, conditions. Examples of foreign matter to consider are dust, sand, mud, rain, snow, ice, hail, condensation, insects, rodents, rodent waste, mould, fungus, rogue mechanical tools, rogue mechanical fasteners, swarf, particles, coffee, cola (which is corrosive), pollen, air-borne seeds, vegetation, water residue, smoke/cordite residue, insect faeces/secretions, snails, worms, brake/clutch dust, hair and textile fibres. Far fetched? No – experience shows that unexpected foreign matter is to be expected.
A New Approach – Direct Measurement
As a general rule, if the position of an object is to be measured accurately then the measurement should be made at, or close to, the object. Measuring shaft angle directly simplifies the system and reduces tolerance stack up. The result: improved accuracy and reliability.
So why doesn’t everyone use direct measurement? The reason is that, until recently, large bore rotary encoders were disproportionately expensive, delicate and difficult to fit. Ring style optical encoders have been around for years but are expensive, bulky, need careful installation and are prone to failure with foreign matter. Similarly, large bore or ‘pancake’ resolvers have been around for years but their price, complex electrical supply/signal processing and bulk make them unsuitable for most mainstream applications.
However, a new generation of inductive encoders now enables simple, effective and accurate angle measurement for large diameter shafts. These devices work on similar principles to contactless resolvers and are just as robust and reliable. Rather than wire spools or windings they use printed, laminar windings. These enable a low profile, annular encoder ideally suited to large through-shafts. Unlike their optical counterparts, these devices do not require precision installation and the electrical interface is straightforward – DC voltage in; absolute digital data out. The mechanical arrangement of these new generation devices is simple and eradicates all gearing. The result: a simple, easy-to-install, compact, lightweight, low inertia, accurate and reliable solution.
About Zettlex UK Ltd:
Zettlex is a sensors company. The company’s range of sensors measure position or speed accurately and reliably, even in harsh conditions.
Securing Industrial Networks in The Connected Enterprise
By Gregory Wilcox, Rockwell Automation
Hackers and malicious individuals are also targeting manufacturers and industrial operators. These...
The Path to Zero Downtime for Remote Assets
By Tom Craven, RRAMAC Connected Systems
Downtime is expensive to both equipment manufacturers and end-users, and amounts to losses greater than...
The story behind the data
By Robert Golightly, AspenTech
The inside story of operational intelligence lies in rich visualization and analytics capabilities that improve...
IoT Impact on Industrial Automation
By Bill Lydon, Editor
The Internet of Things (IoT) is creating a flood of new technology. As a result, the automation industry is fortunate to...
Choosing the right switchboard
By Pat McLaughlin, Boulting Technology
Switchboards generally have relatively long lives and can last anywhere between 25 and 40 years. Here’s...
Halter CNC robotic loading arm feeds anchor bolt milling machines
Since its installation, the robotic loading arm has increased output from its assigned CNC milling machine by at least 150 leaf chain anchor bolts...
H2W offers linear motion calculators
Calculators determine how much force and acceleration are necessary to achieve the desired motion profile, and even in selecting which is the best...
RIA appoints Carole Franklin Director of Standards Development
Franklin is an expert project manager and facilitator whose work with consulting firm Booz Allen Hamilton included process improvement and...
Rockwell Automation updates Integrated Architecture Portfolio
The portfolio includes an Allen-Bradley controller, graphic terminal, servo drive and distributed I/O system, as well as the latest release of the...
Automation Fair show report
More than 18,000 people attended the 2015 Automation Fair in Chicago, IL.