ECD introduces DCA-23 Seawater Dechlorination Analyzer
March 10, 2011 -- The new Model DCA-23 Seawater Dechlorination Analyzer from Electro-Chemical Devices (ECD) is a low maintenance measurement solution that requires no chemicals to monitor seawater chlorine levels accurately in a wide range of industrial process and municipal water treatment applications.
Seawater has multiple process industry uses, including heat exchange systems at LNG terminals, oil/gas refineries, electric power plants and SWAC systems (salt water air conditioning) in commercial buildings. Seawater desalination plants also chlorinate and dechlorinate the water before processing. The chlorination of seawater inhibits the growth of marine life on the various filters, screens and heat exchanger surfaces, but the seawater must be dechlorinated for environmental reasons before it is discharged back into the original water supply.
The DCA-23 is a single-purpose analyzer that monitors chlorine in seawater from 0.0 to 2.00 ppm. It features an economical, low-maintenance design that requires no chemicals. It measures the pH, temperature and Total Residual Oxidant (TRO) of seawater and converts them into an equivalent ppm chlorine value, which is displayed to conform with existing conventions. It features a minimum detection limit (MDL) of 0.014 ppm and a limit of quantification (LOQ) of 0.044 ppm, making the DCA-23 Analyzer suitable for monitoring in environments with the most stringent dechlorination regulations.
The DCA-23 features three sensors, pH, TRO and temperature. The pH sensor measures from 0-14 pH, the TRO sensor measures from ±1500 mV and the temperature sensor measures from 0-100°C. The DCA-23 with its tri-parameter monitoring, is the ideal seawater chlorine analyzer for a wide range of applications.
During the re-gasification process at LNG terminals, large quantities of chlorine-treated seawater are used to heat the LNG heat exchangers. Seawater gradually raises the temperature of cold liquid natural gas changing it from a compressed fluid into a gas for distribution through the service pipelines. Recycling the seawater requires dechlorination monitoring with the DCA-23.
Steam-driven turbines at electric power plants generate much of the world’s electricity. Seawater cooling is employed at both thermal and nuclear power plants in coastal regions. The seawater flows over heat exchangers and condenses the steam used to drive the turbines. The cooling is accomplished with chlorinated seawater, which requires monitoring with the DCA-23 before discharge. Similar heat exchange processes are utilized by oil/gas refineries.
In water desalination plants, seawater is treated by reverse osmosis (RO) to remove the salt. The intake seawater is treated with chlorine to kill algae and other organisms prior to the reverse osmosis process. The RO membranes are degraded by chlorine so the chlorinated seawater must be dechlorinated before contacting the membranes. The DCA-23 is ideal for monitoring the dechlorination process.
The DCA-23 Seawater Dechlorination Analyzer is available in five configurations: 1.) Loop powered, 4-20 mA = 0.00 to 2.00 ppm Cl; 2.) 24VDC powered with two 4-20 mA outputs, Cl and pH; 3.) 24VCD powered with two 4-20 mA outputs and two alarm relays; 4. ) 110 VAC powered with two 4-20 mA outputs and two alarm relays; and 5.) 220 VAC powered with two 4-20 mA outputs and two alarm relays. The maximum system pressure and temperature rating is 50 psig at 60°C.
The DCA-23 Seawater Dechlorination Analyzer is an easy to install plumb-and-play device when ordered as a complete Uni-Strut rack mounted system and it is also available as separate components. The rack mounted system is plumbed with ¾” schedule 80 pipe, with slip fittings for input and drain. The T-23 Analyzer, 2 sensors, pH flow cell and TRO flow cell are rack mounted using 1⅝”x1⅝” Uni-Strut rail.
About Electro-Chemical Devices
Electro-Chemical Devices (ECD) is a manufacturer of liquid analytical process instrumentation. Founded in 1977, ECD has been in business for over 30 years.
Omron’ Mode E3Z-T61 photoelectric sensor
Omron’ Mode E3Z-T61 photoelectric sensor have a top-level sensing distance, and allows sensing distance to be as long as 30 m
15.6” Panel Mount Monitor Built of Aluminum Die-Casting Chassis with IP65 Front Bezel
Resistive type touch screen monitor displays 1366 x 768 resolution, supports VGA and DVI video input, and is panel mount or VESA mount ready.
Comau introduces Racer5-0.63 & Racer5-0.80 industrial robots
Racer5-0.63 and Racer5-0.80 are both controlled by the R1C 19" rack-mounted controller, which can be integrated into a single cabinet to control an...
Knick introduces P50000 line of transducers
The ProLine P50000 series from Knick was specially developed for voltage and current measurement in main and auxiliary power circuits of...
FLIR releases upgraded A35/A65 temperature sensors
Designed for process control and quality assurance, the cameras are available with 10 field-of-view options, from 8 degrees up to 90 degrees, to...
MCAA appoints Teresa Sebring as President of the Association
Mrs. Sebring joined MCAA in 1997 as Administrative Assistant to the President. Since 2012 she has served as Vice-‐President and has been...
NI announces Alex Davern to succeed Dr. James Truchard as CEO
Davern will succeed Dr. James Truchard, age 73, who has served as the Chief Executive Officer of NI since the company's founding in 1976, effective...
Flowserve Collaborates with NI, HP and PTC to Enable Predictive Maintenance for Process Control
By Cory Fogg, Automation.com
Efficient predictive maintenance has been a goal of the process industry for ages. This article discusses the efforts...
2016 ISA Process Control & Safety Symposium and Exhibition features three acclaimed speakers
Attending engineers, technicians and plant managers will be able to gain the latest insights, news and trends in process measurement and control,...
Adding Wireless Instruments to Existing Networks
By Moazzam Shamsi, Emerson Process Management
Once a wireless infrastructure is in place, new instruments can be added quickly and easily by...